Manuals and User Guides for Katronic Technologies FLUXUS ADM We have 1 Katronic Technologies FLUXUS ADM manual available for free PDF . Wall thickness measurement (option). FLUXUS ADM FLUXUS ADM Portable Ultrasonic Flow Measurement of. Liquids. Features. Rent, lease or buy the Ultrasonic Flowmeter (Flexim Fluxus ADM ) from Livingston. Request the rental price for the Ultrasonic Flowmeter (Flexim Fluxus.
|Published (Last):||9 March 2010|
|PDF File Size:||16.36 Mb|
|ePub File Size:||15.2 Mb|
|Price:||Free* [*Free Regsitration Required]|
Set-up Of The Storage Interval Set-up Of The Measuring Quantity Selection Of The Measuring Point Overview Of The Wtm Option Deletion Of Measured Values Utility Program Flxtotxt Alarm Outputs During Transducer Positioning Measuring With Calculation Channels Disturbances During Measurement Printing Of Measured Values Parameterisation Of The Calculation Channels ffluxus Units Of Measurement Integrated Measuring Point Multiplexer Super Dluxus Mode The status display enables even 67225 inexperienced user to judge online the quality and precision of the measurement.
Deletion Of Parameter Records Transducer Installation In Reflection Mode Set-up Of Analogue Outputs Program Branch Mea Overview Malfunctions During Super User Mode Set-up Of Voltage Output The clamp-on flow transducers are simply mounted onto the pipe from the outside.
Wtm Program Branch Parameter Undisturbed Flow Profile Program Branch Opt Overview Cleaning Of Outside Pipewall 66725 With its clamp-on transducers and its rechargeable battery, it is an ideal tool for service work. Instrument Start-up Basics Types Of Display Templates Recognition Of Flow Direction Input Of The Correction Offsets The user interface is adapted to the detected transducers.
Heat Quantity Measurement optional Change To Operational Mode Wtm Don’t have an account? Program Branch Measuring The transducers are not in direct contact with the medium to be measured and thus in no way submitted to pressure and wear and tear.
Operational Modes To Set-up Set-up Of Alarm Outputs Errors During Parameter Input Total Quantity Storage Program Branch Parameter Transfer Of Measured Values Measurement Is Not Successful Process Interfaces optional Start Of The Measurement Wall Thickness Measurement optional Utility Program Movefast Measurement Of The Sound Velocity Heat Quantity Measurement hqm Set-up Of Other Output Options Fluxxus Of Flowmeter Set-up Of Alarm Properties With Tension Strap Input Of Medium Parameters Identification Of The Measuring Point Stop Wall Thickness Measurement